Magnetic resonance (MR) thermometry is a valuable method for characterizing thermal fields generated by high intensity focused ultrasound (HIFU) transducers in tissue phantoms and excised tissues. However, infiltration of noise signals generated by external rf sources into the scanner orifice limits the ability of the scanner to measure temperature rise during the heating or ablation phase. In this study, magnetic resonance interferometry (MRI) monitored HIFU ablations are performed on freshly excised porcine liver samples, at varying sonication times, 20 s, 30 s, and 40 s at a constant acoustic intensity level of $1244 W/cm2$. Temperature throughout the procedure was measured using proton resonant frequency MR thermometry. Without filtering, reliable temperature measurements during the heating phase could not be obtained since temperature maps appeared blurred and analysis was impossible. Also, measurements acquired during the cooling phase decayed manifested an unrealistically slow rate of temperature decay. This abnormally slow rate was confirmed with computational results. A low-pass $RC$ filter circuit was subsequently incorporated into the experimental setup to prevent infiltration of noise signals in the MRI orifice. This modified $RC$ filter circuit allowed noninvasive measurement of the HIFU induced temperature rise during the heating phase followed by temperature decay during cooling. The measured data were within 13% agreement with the temperature rise computed by solving the acoustic and heat equations.

1.
Qian
,
Z. W.
,
Xiong
,
L.
,
Yu
,
J.
,
Shao
,
D.
,
Zhu
,
H.
, and
Wu
,
X.
, 2006, “
Non Invasive Thermometer for HIFU and Its Scaling
,”
Ultrasonics
0041-624X,
44
, pp.
e31
e35
.
2.
Oshiro
,
T.
,
Sinha
,
U.
,
Lu
,
D.
, and
Sinha
,
S.
, 2002, “
Reduction of Electronic Noise From Radiofrequency Generator During Radiofrequency Ablation in Interventional MRI
,”
Journal of Computer Assisted Tomography
0363-8715,
26
, pp.
308
316
.
3.
Dasgupta
,
S.
,
Wansapura
,
J.
,
Hariharan
,
P.
,
Pratt
,
R.
,
Witte
,
D.
,
Myers
,
M. R.
, and
Banerjee
,
R. K.
, 2010, “
HIFU Lesion Volume as a Function of Sonication Time, as Determined by MRI, Histology, and Computations
,”
ASME J. Biomech. Eng.
0148-0731,
132
(
8
), p.
081005
.
4.
Rieke
,
V.
,
Vigen
,
K.
,
Sommer
,
G.
,
Daniel
,
B. L.
,
Pauly
,
J. M.
, and
Butts
,
K.
, 2004, “
Referenceless PRF Shift Thermometry
,”
Magn. Reson. Med.
0740-3194,
51
, pp.
1223
1231
.
5.
Hariharan
,
P.
,
Myers
,
M. R.
, and
Banerjee
,
R. K.
, 2007, “
HIFU Procedures at Moderate Intensities—Effect of Large Blood Vessels
,”
Phys. Med. Biol.
0031-9155,
52
, pp.
3493
3513
.
6.
Khokhlova
,
V. A.
,
Bailey
,
M. R.
,
Reed
,
J. A.
,
Cunitz
,
B. W.
, and
Kaczkowski
,
P. J.
, 2006, “
Effects of Nonlinear Propagation, Cavitation, and Boiling in Lesion Formation by High Intensity Focused Ultrasound in a Gel Phantom
,”
J. Acoust. Soc. Am.
0001-4966,
119
, pp.
1834
1848
.
You do not currently have access to this content.