The Medical Device Amendments to the Food, Drug, and Cosmetics Act mandates that substantially new devices be subjected to study and testing prior to the commencement of clinical trials. The proposed outline for preclinical analysis, testing, and evaluation of orthopaedic implants is intended to guide the designer in defining what the limits of understanding are and, within those limits, to minimize the risk to implant recipient. The process of design qualification and verification is broken down into seven major steps: Problem Definition, Underlying Assumptions, Material Selection, Engineering Analysis, Correlation of Demand to Capabilities, Analytical Estimate of Performance and Experimental Verification of Performance. The discussion of each of these topics focuses on the type of information needed for each activity. Thus, these guidelines present a framework for the rational innovation and design of orthopaedic implants which minimizes the risk to patients involved in clinical trials.
Preclinical Evaluation of Orthopaedic Implants
J. H. Butler,
University of Florida, Gainesville, Fla.
J. H. Butler
Oak Ridge National Laboratories, Oak Ridge, Tenn.
D. C. Richardson
Food and Drug Administration, Silver Spring, Md.
- Views Icon Views
- Share Icon Share
- Cite Icon Cite
- Search Site
Piotrowski, G., Butler, J. H., and Richardson, D. C. (October 1, 1980). "Preclinical Evaluation of Orthopaedic Implants." ASME. J. Mech. Des. October 1980; 102(4): 655–662. https://doi.org/10.1115/1.3254802
Download citation file:
- Ris (Zotero)
- Reference Manager
Get Email Alerts
When, Why, and How Can Derivative-Free Optimization Be Useful to Computational Engineering Design?
J. Mech. Des (January 2020)
A Machine Learning Enabled Multi-Fidelity Platform for the Integrated Design of Aircraft Systems
J. Mech. Des (December 2019)