The time-dependent relative entropy field at the impeller exit of a centrifugal compressor is measured. This study is part of a broader effort to develop comprehensive measurement techniques that can be applied in the harsh environment of turbomachines. A miniature unsteady entropy probe (diameter of 1.8 mm) is designed and constructed in the present study. The unsteady entropy probe has two components: a one-sensor fast-response aerodynamic probe and a pair of thin-film gauges. The time-dependent total pressure and total temperature are measured from the fast-response aerodynamic probe and pair of thin-film gauges, respectively. The time-dependent relative entropy derived from these two measurements has a bandwidth of 40 kHz and an uncertainty of ±2 J/kg. The measurements show that for operating Condition A, $φ$=0.059 and $ψ$=0.478, the impeller exit flowfield is highly three dimensional. Adjacent to the shroud there are high levels of relative entropy and at the midspan there are low and moderate levels. Independent measurements made with a two-sensor aerodynamic probe show that the high velocity of the flow relative to the casing is responsible for the high relative entropy levels at the shroud. On the other hand, at the midspan, a loss free, jet flow region and a channel wake flow of moderate mixing characterize the flowfield. At both the shroud and midspan, there are strong circumferential variations in the relative entropy. These circumferential variations are much reduced when the centrifugal compressor is operated at operating Condition B, $φ$=0.0365 and $ψ$=0.54, near the onset of stall. In this condition, the impeller exit flowfield is less highly skewed; however, the time-averaged relative entropy is higher than at the operating Condition A. The relative entropy measurements with the unsteady entropy probe are thus complementary to other measurements, and more clearly document the losses in the centrifugal compressor.

1.
Denton
,
J. D.
, 1993, “
Loss Mechanisms in Turbomachines
,”
ASME J. Turbomach.
0889-504X,
115
, pp.
521
652
.
2.
Inoue
,
M.
, and
Cumpsty
,
N. A.
, 1984, “
Experimental Study of a Centrifugal Impeller Discharge Flow in a Vaneless and Vaned Diffusers
,”
ASME J. Eng. Gas Turbines Power
0742-4795,
106
, pp.
455
467
.
3.
Roduner
,
C.
,
Köppel
,
P.
,
Kupferschmied
,
P.
, and
Gyarmathy
,
G.
, 1999, “
Comparison of Measurement Data at the Impeller Exit of a Centrifugal Compressor Measured With Both Pneumatic and Fast-Response Probes
,”
ASME J. Turbomach.
0889-504X,
121
, pp.
609
618
.
4.
Strahlecker
,
D.
, and
Gyarmathy
,
G.
, 1998, “
Investigations of Turbulent Flow in a Centrifugal Compressor Vaned Diffuser by 3-Component Laser Velocimetry
,” ASME Paper No. 98-GT-300.
5.
Schleer
,
M.
, and
Abhari
,
R. S.
, 2006, “
Clearance Effects on the Evolution of the Flow in the Vaneless Diffuser of a Centrifugal Compressor at Part Load Condition
,” ASME Paper No. GT2006–90083.
6.
Pfau
,
A.
,
Schlienger
,
J.
,
Kalfas
,
A. I.
, and
Abhari
,
R. S.
, 2003, “
Unsteady 3-Dimensional Flow Measurement Using a Miniature Virtual 4 Sensor Fast Response Aerodynamic Probe (FRAP)
,” ASME Paper No. GT2003–38128.
7.
Schmidt
,
E.
, and
Werner
,
K.
, 1941, “
Wärmeabgabe über den Umfang eines Angebblasenen Geheizten Zylinders
,”
Forsch. Ingenieurwes.
0015-7899,
12
, pp.
65
73
.
8.
Hartunian
,
R. A.
, and
Varwig
,
R. I.
, 1962, “
On Thin-Film Heat-Transfer Measurements in Shock Tubes and Shock Tunnels
,”
Phys. Fluids
0031-9171,
5
, pp.
169
174
.
9.
1970,
Thermophysical Properties of Matter
,
The TPRC Data Series: Thermal Conductivity Nonmetallic Solids
Vol.
2
,
Y. S.
Touloukian
and
C. Y.
Ho
, eds.,
IFI/Plenum
,
New York
;
Specific Heat Nonmetallic Solids
, Vol.
5
.
10.
Lomas
,
C. G.
, 1986,
Fundamentals of Hot Wire Anemometry
,
Cambridge University Press
,
London
.
11.
Skinner
,
G. T.
, 1960, “
Analogue Network to Convert Surface Temperature to Heat Flux
,”
American Rocket Society Journal
,
30
, pp.
569
570
.
12.
Schultz
,
D. L.
, and
Jones
,
T. V.
, 1973, “
Heat Transfer Measurements in Short Duration Facility Hypersonic Facilities
,” Agardograph No. 165.
13.
Oldfield
,
M. L. G.
,
Burd
,
H. J.
, and
Doe
,
N. G.
, 1982, “
Design of Wide Bandwidth Analogue Circuits for Heat Transfer Instrumentation in Transient Wind Tunnels
,”
16th Symposium of International Center for Heat and Mass Transfer
,
Hemisphere
,
Washington, DC
, pp.
233
257
.
14.
Epstein
,
A. H.
,
Guenette
,
G. R.
,
Norton
,
R. J. G.
, and
Yuzhang
,
C.
, 1986, “
High-Frequency Response Heat-Flux Gauge
,”
Rev. Sci. Instrum.
0034-6748,
57
, pp.
639
649
.
15.
Kupferschmied
,
P.
, 1998, Zur Methodik Zeitaufgelöstter Messungen mit Strömungssonden in Verdichtern und Turbinen, Diss ETH No. 12474.
16.
Hunziker
,
R.
, and
Gyrmathy
,
G.
, 1993, “
The Operational Stability of a Centrifugal Compressor and Its Dependence on the Characteristics of the Subcomponents
,” ASME Paper 93-GT-284.
17.
Schleer
,
M. W.
, 2006, “
Flow Structure and Stability of a Turbocharger Centrifugal Compressor
,” Diss ETH No. 16605.
18.
Schleer
,
M.
,
Mokulys
,
T.
, and
Abhari
,
R. S.
, 2003, “
Design of a High Pressure-Ratio Centrifugal Compressor for Studying Reynolds Number Effects
,”
International Conference on Compressors and their Systems
,
London
, pp.
391
404
.
19.
Dean
,
R. C.
, and
Senoo
,
Y.
, 1960, “
Rotating Wakes in Vaneless Diffusers
,”
ASME J. Basic Eng.
0021-9223,
82
, pp.
563
574
.